Standard Practice for Continuous Sizing and Counting of Airborne Particles in Dust-Controlled Areas and Clean Rooms Using Instruments Capable of Detecting Single Sub-Micrometre and Larger Particles
适用范围：<p>The primary purpose of this practice is to describe a procedure for collecting near real-time data on airborne particle concentration and size distribution in clean areas as indicated by single particle counting techniques. Implementation of some government and industry specifications requires acquisition of particle size and concentration data using an SPC.</p> <p>The processing requirements of many products manufactured in a clean room involves environmental cleanliness levels so low that a single particle counter with capability for detecting very small particles is required to characterize clean room air. Real-time information on concentration of airborne particles in size ranges from less than 0.1 <span class="unicode">x03BC;</span>m to 5 <span class="unicode">x03BC;</span>m and greater can be obtained only with an SPC. Definition of particles larger than approximately 0.05 <span class="unicode">x03BC;</span>m may be carried out with direct measurement of light scattering from individual particles; other techniques may be required for smaller particles, such as preliminary growth by condensation before particle measurement.</p> <p>Particle size data are referenced to the particle system used to calibrate the SPC. Differences in detection, electronic and sample handling systems among the various SPCs may contribute to differences in particle characterization. Care must be exercised in attempting to compare data from particles that vary significantly in composition or shape from the calibration base material. Variations may also occur between instruments using similar particle sensing systems with different operating parameters. These effects should be recognized and minimized by using standard methods for SPC calibration and operation.</p> <p>In applying this practice, the fundamental assumption is made that the particles in the sample passing through the SPC are representative of the particles in the entire dust-controlled area being analyzed. Care is required that good sampling procedures are used and that no artifacts are produced at any point in the sample handling and analysis process; these precautions are necessary both in verification and in operation of the SPC.</p><p id="s00002">1.1 This practice covers the determination of the particle concentration, by number, and the size distribution of airborne particles in dust-controlled areas and clean rooms, for particles in the size range of approximately 0.01 to 5.0 <span class="unicode">x03BC;</span>m. Particle concentrations not exceeding 3.5 <span class="unicode">x00D7;</span> 10<sup>6</sup> particles/m<sup>3</sup> (100 000/ft<sup>3</sup>) are covered for all particles equal to and larger than the minimum size measured.</p> <p id="s00003">1.2 This practice uses an airborne single particle counting device (SPC) whose operation is based on measuring the signal produced by an individual particle passing through the sensing zone. The signal must be directly or indirectly related to particle size. <p class="desc" id="N00001"> <span class="smallcap">Note</span> 18212;The SPC type is not specified here. The SPC can be a conventional optical particle counter (OPC), an aerodynamic particle sizer, a condensation nucleus counter (CNC) operating in conjunction with a diffusion battery or differential mobility analyzer, or any other device capable of counting and sizing single particles in the size range of concern and of sampling in a cleanroom environment.</p> </p> <p id="s00004">1.3 Individuals performing tests in accordance with this practice shall be trained in use of the SPC and shall understand its operation.</p> <p id="s00005">1.4 Since the concentration and the particle size distribution of airborne particles are subject to continuous variations, the choice of sampling probe configuration, locations and sampling times will affect sampling results. Further, the differences in the physical measurement, electronic and sample handling systems between the vario......
标准组织： ASTM - 美国材料与试验协会标准
英文关键词： air sampling airborne particles particle counting particle size distribution particles Atmospheric analysis Calibration--semiconductor analysis instrumentation Cleanrooms (materials/applications) Counting airborne particles Dust-controlled areas Light scattering Particle counters/counting Particulate contamination (aerospace environments) Pulse height Size/size distribution analysis
网络出版服务许可证：(署)网出证(京)字第072号 药品医疗器械网络信息服务备案：(京)网药械信息备字（2023）第 00470 号 信息网络传播视听节目许可证 许可证号：0108284